Frequently Asked Questions
Here are a few solutions to frequently asked questions we’ve received from our customers. If you do not see a solution to a problem you’re currently dealing with, please let us know. We may need to add it to this list.
Each PowerMaster unit that is shipped contains a CD which has the user manual and all related software that is required. Additionally, there are videos on the website to help get you up and running.
Verify the ground and neutral are connected correctly. Verify in the setup that the correct form and service are selected.
Verify your operating system and download the appropriate file from the download section of the website. Inside the zip file is an instruction manual that walks you through the process.
Why does one CT fail the ratio test while another passes, even though they both have the same ratio error?
CT test results are dependent on the IEEE parallelogram test method. The pass/fail points are plotted with ratio being your “x” component and phase error being the “y” component. For further explanation on CT testing and results, please download our app note on CT testing.
The fundamental is not shown on the harmonic graph, which allows the other harmonics to be displayed in bar format in a more user-friendly configuration.
Yes, the master database can be located anywhere your PC can write to. Launch meter site manager, click on File > Change Master Database Directory. Browse to the location and select OK.
In Meter Site Manager, how can I print all the reports associated with a site test at the same time?
Select the time stamp of the test you wish to print, the use Ctrl-F1 to print all.